Batteries should be changed when the battery voltage falls below 7.6 volts.
There is a negligible effect of decreased battery voltage, down to the minimum battery voltage of 7.6V. This is due to the use of linear voltage regulators on the DMMCheck Plus.
Yes, please wait 10 minutes prior to use after turning on your DMMCheck Plus. For the highest accuracy, the ambient temperature should be as close as possible to the calibration temperature noted on the unit (typically 78 degrees F).
Yes, the aging rate of the LT1021 voltage reference IC, used in the DMMCheck Plus, is proportional to the square root of the number of powered-on kilo-hours. For example, if, say, the voltage gradually drifts 10uV in 1000 hours, then the drift in the next 4000 hours would be expected to be 20uV.
Yes, we accept international orders. If you do not see your country on the list of ship-to countries, please contact us. International shipments may be subject to additional shipping costs to ensure safe and expedient delivery. FedEx shipping is available upon request. EU customers may also order through Welecton, detailed further in the "INTERNATIONAL CUSTOMERS" tab.
The LT1021BCN8-5 voltage reference IC, used in this product, has a worst-case load regulation value of 20ppm/mA. This means that for every 1mA of load current, the output can change by up to 100uV. Our testing indicates that this specification is conservative and most IC's are at least twice as good as this worst-case number. This load regulation specification is good to loads up to 10mA. We recommend a load of 1 MΩ or higher for best accuracy.